I have a 2kN load cell. I took it apart to try and understanding how the wires connect to the strain gauge and how the load is read on the serial pad.
I am curious how the loads are read on the strain gauge. There are 4 strain gauges on the base and it appears there are strain gauges on the side as well.
Can anyone help explain how this works?
Also, is there a way I can add pictures to help explain what I am talking about?
February 26, 2019
1/ A single strain gauge is simply a thin plastic base with a conductor fastened to it in a zig zag pattern arranged so that when the base is stretched the conductor is stretched. This changes the conductor resistance. 2/ In a load cell 4 strain gauges are glued to the metal (usually aluminum) load cell base at the parts which are stretched or compressed under load and so the strain gauge resistance is increased or decreased under load. 3/ The 4 strain gauges are connected in a Wheatstone bridge configuration (Google or Wickipedia "Wheatstone Bridge" for a deeper analysis of this). 4/ The bridge is supplied with a stable voltage at the excitation input and generates a very small voltage at the signal output which must be amplified and this amplified voltage goes to an A/D converter. 5/ Take a look at " https://www.sparkfun.com/products/13879 " for a more detailed description of the strain gauge amplifier.
February 26, 2019
Anyone can ask a question.
Did you already search (see above) to see if a similar question has already been answered? If you can't find the answer, you may ask a question.
CircuitLab's Q&A site is a FREE questions and answers forum for electronics and electrical engineering students, hobbyists, and professionals.
We encourage you to use our built-in schematic & simulation software to add more detail to your questions and answers.
Please respect that there are both seasoned experts and total newbies here: please be nice, be constructive, and be specific!
CircuitLab is an in-browser schematic capture and circuit simulation software tool to help you rapidly design and analyze analog and digital electronics systems.
|New @ CircuitLab|